Salt spray test chamber parts, calibration, & service
A discussion started in 1999 but continuing through 2020
Spare parts for "Harshaw Chemical" salt spray environmental chamber?1999
Q. We are looking for some spare parts for an "environmental chamber - metal finishing equipment" manufactured by "the Harshaw Chemical Co." in Cleveland, Ohio. We cannot get information about that company and we need to repair the environmental chamber.
Model: 21 Serial:
If somebody can help us we appreciate it. We need information as soon as possible.Jose Ignacio Martinez
- Miami, Florida
A. Hi Jose. Although Harshaw Chemical Co. is now part of Atotech, the Equipment Division, which manufactured the salt spray cabinets, was spun off as Auto Technology in Strongsville, Ohio. That's probably who you're looking for :-)
Ted Mooney, P.E.
finishing.com - Pine Beach, New Jersey
Striving to live Aloha
Salt spray cabinet qualification requirements?May 13, 2011
Q. Is there a published specification for calibration or qualification requirements for Salt Spray Cabinet and Humidity Cabinet equipments? We need information for requirements as per ASTM Standard or ISO Standards Or any other standard.
Example: Salt Spray Cabinet equipment or Humidity test chamber unit must be calibrated once a year or once every 2 years.
Metal Finishing, Chemist - Columbia, South Carolina, U.S.A.
May 20, 2011
A. ASTM B117 [affil. link to spec at Techstreet]
electroplater - Galva, Illinois
June 24, 2011
A. Usually we recommend our customers to calibrate their salt spray chambers every year.Davey Shaw
Shenzhen, Guangdong, China
July 11, 2012
Q. We recently purchased a salt spray chamber and we are working to get a certification of the ISO 17025 [affil. link to spec at Techstreet] for the laboratory.
Does this certification standard require a calibration of the pH meter, hydrometer and thermometer?
In addition, do the collectors have to be on the same level as the samples?
Finally, How do you clean the chamber to eliminate the rust?
Thanks a lot for your help.
- Montreal, Canada
Salt spray chamber repairs?January 21, 2015
Q. I have an Auto Technology chamber and it will stop fogging after 2 hours. We have done everything in our power to have this gone through and looked at by an engineer. I have tested the solution, installed a rebuild kit and everything. All and any advice would be great.
RFQ: Does anyone know who is in the WI area that works on these chambers? For the cost of having the Auto Tech come to WI I can buy a new chamber.Brian Arneson
QualityTech - Oshkosh Wisconsin USA
Ed. note: Brian and other readers may also be interested in letter no. 50132, "Salt Spray Test -- ASTM B117 Operational Problems".
March 18, 2015
A. Was a new atomizer included in the rebuild kit? They're difficult to keep clean.
Check your bubble tower: does it have buildup on the walls?
Dilute HCl cleans the buildup off nicely. Rinse it off well.
The bubbler at the bottom of the tower could be clogged as well.
Check all air fittings to ensure no leaks.
- Seattle, Washington USA
March 19, 2015
A. Have you tried looking for debris in the atomizer? I had a similar issue a few weeks ago and it ended up being a spec of something was blocking the inlet for salt water.Dan Harris
- Fort Walton Beach, Florida, USA
March 6, 2019
RFQ: We have the same unit and can't find anyone to repair it due to the age. I would really appreciate any contacts for repair of a chamber.K Johnson
- Birmingham, Alabama
^- Privately contact this inquirer -^
A. Hi K. Decades ago I worked for Harshaw Equipment Division which built those cabinets (now Auto Technology), and when we repaired PVC-lined steel salt spray chambers it was often a very expensive job because, after decades of very corrosive solutions leaking under the lining, after stripping the lining and trying to rebuild the shell, there often wasn't much worth saving. So good luck if you have a very old PVC-lined steel one :-)
Ted Mooney, P.E. RET
finishing.com - Pine Beach, New Jersey
Aloha -- an idea worth spreading
January 6, 2020
A. We had an old model 21 Harshaw chamber that rusted out. We had a new enclosure made out of 3/8 thick polypropylene sheet heat welded together. It was built just like the original design with the double walled water jacket. we keep all the original hardware and controls. We have been running it continuously for probably 15 years now with no problems.Dave Crabb
- Grand Junction, Colorado, US