Surface roughness conversion
Q. I always receive the surface roughness measurement in Rmax but our company use Ra method. Are there any conversion factor to convert Rmax to Ra or vice versa?
I find all the handbooks in our library but no answer.
Would anybody tell me the conversion factor I'd very appreciate.Agnes Luk
IOU-AL - HK
You should visit www.predev.com/smg/ Surface Metrology Guide
Good Luck,Bill Boatright
Houghton International - Raleigh, North Carolina
Q. I'm familiar with Ra measurements, but is anyone familiar with SRa measurements, that could give me the definition of that abbreviation? Thanks.Jim Sherburne
Stamping Engineer - Smyrna, Tennessee, USA
February 3, 2009
Q. What is the definition of Rmax?
What is the difference between Ry Din and Rmax JIS.
EMPLOYEE - INDIA
June 18, 2009
Q. I want conversion of following surface finish in Ra
It is shown on Japanese drawings and part material is SAE52100 (Bearing steel)
Bearing Industry - Thane, Maharashtra, India
June 10, 2009
A. Hi, Vasant. The main reason there are so many different surface measurement standards is that they all measure different things that may be important for different reasons.
Consider for example, that the anticipated life of a polymer shaft seal might depend on the average roughness, whereas an electroplating shop knows that plating cannot fill scratches, it accentuates them -- so a plating shop uses a roughness scale that talks about the deepest scratch, not the average.
You can't actually convert a value in one scale to the other because they measure things that tend to "track each other" but aren't the same thing, like torque vs. horsepower, or people's height vs. weight. The average weight of people who are 6 foot tall is obviously greater than the weight of people who are 5 foot tall, but you can't covert height to weight. These surface finish "conversions" are rough estimates that include a large factor of safety (which proves expensive and always works against you, and still may not be 100% safe). The best approach is to obtain an instrument that measures what you are trying to measure.
But good explanations, pictorials, and 'conversion' information can be found at www.misumi.co.th/products/technical/press/pdf/pr1167_1168.pdf
Ted Mooney, P.E. RET
Pine Beach, New Jersey
This public forum has 60,000 threads. If you have a question in mind which seems off topic to this thread, you might prefer to Search the Site