Boric acid determination, autotitration
Currently I have some problems with autotitration of boric acid in nickel plating bath, autotitrator is MT 70ES. I add mannitol into sample and use NaOH as titrant, with pH electrode to measure endpoint. In last three months, I found signal at endpoint became lower and lower, dropped from about 2000 mv/ml to 100 mv/ml, meanwhile endpoint became noise and results were not so repeatable. After evaluation, I did not find any problem with autotitrator and chemicals, for new prepared nickel bath, signal at endpoint is still about 2000 mv/ml.
Q1: why does signal drop so much with extension of nickel bath life?
Q2: Is there any way to recover endpoint signal to high level, eg. 1000 mv/ml?Wang Li Ping
My guess is that you have drag in and or breakdown products that are forming a buffer that interferes with your titration.
I used color indicators, but the result is pH sensitive so are somewhat equal. I never saw the phenomenon you speak of.James Watts
- Navarre, Florida
This public forum has 60,000 threads. If you have a question in mind which seems off topic to this thread, you might prefer to Search the Site